We offer the widest product range of low-flow (mass) flow meters and controllers on the market. Numerous styles of both standard and bespoke instruments can be offered for applications in laboratory, machinery, industry and hazardous areas.
Customer satisfaction, innovation and quality of product and service have been the cornerstones of Bronkhorst's success. Based on our experience, innovation and sense of responsibility, a relationship with Bronkhorst assures Performance for Life.
Bronkhorst instruments are used for numerous applications in many different markets. In this section you will find an overview of the main markets for our equipment, illustrated with some typical examples of applications.
Are you looking for technical documentation, are you interested to learn more about the measuring principles of Bronkhorst products, or you do want to get in contact with a Bronkhorst Service Engineer? This section will guide you to the relevant service & support topics.
Bronkhorst High-Tech BV the leaders in Mass Flow Meter / Mass Flow Controller technology for gases and liquids, Pressure Controllers and Evaporation Systems.
You would think that measurements of mass flow would be expressed in units of mass, such as grams/hour, milligrams/second etc. Most users, however, think and work in units of volume.
According to this example, mass flow should actually be expressed in units of weight such as g/h and mg/s. Many users, however, think and work in units of volume. This not a problem, provided conditions under which the mass is converted to volume are agreed upon.
In order to use density in converting mass flow to volumetric flow, we must pick a set of specific pressure and temperature conditions at which we use the density value for the gas.
The conditions that are agreed upon contain various references, normal reference and standard reference, available in European or American style. What is de difference?
Following the ‘European’ definition, a temperature of 0°C and a pressure of 1,013 bar are selected as ‘normal’ reference conditions, indicated by the underlying letter “n” in the unit of volume used (mln/min or m3n/h). The direct thermal mass flow measurement method is always based on these reference conditions unless otherwise requested.
An example conversion to volumetric units using Normal reference conditions: The mass flow meter indicates 100 g/h of Air flow. • Density Air (@ 0°C) = 1.293 kg/m3 • X ln/m Air = 100 g/h / (60 minutes x 1.293 kg/m3) • Flow = 1.29 ln/m Air
NL-7261 AK Ruurlo (NL)
Tel. +31 573 458800