We offer the widest product range of low-flow (mass) flow meters and controllers on the market. Numerous styles of both standard and bespoke instruments can be offered for applications in laboratory, machinery, industry and hazardous areas.
Customer satisfaction, innovation and quality of product and service have been the cornerstones of Bronkhorst's success. Based on our experience, innovation and sense of responsibility, a relationship with Bronkhorst assures Performance for Life.
Learn more about how Coriolis-based flow meters can be used as microfluidic flow meter instead of thermal flow meters. Read the customer story of a microfluidics system builder who used Bronkhorst products.
Are you looking for technical documentation, are you interested to learn more about the measuring principles of Bronkhorst products, or you do want to get in contact with a Bronkhorst Service Engineer? This section will guide you to the relevant service & support topics.
Bronkhorst High-Tech BV the leaders in Mass Flow Meter / Mass Flow Controller technology for gases and liquids, Pressure Controllers and Evaporation Systems.
This week we have a guest blog from Dr. Roland Snijder, Medical Physicist Resident at Haaglanden Medisch Centrum (NL). To obtain his PhD degree at the Utrecht University, Roland worked as a researcher on the multi-infusion project at the department of Medical Technology & Clinical Physics of University Medical Center Utrecht (UMC Utrecht). His research focused on investigating physical causes of dosing errors in multi-infusion systems. In this research flow characteristics of multi-infusion setups were investigated using Bronkhorst Coriolis flow meters. In this blog Roland explains more about his research.
Often patients in critical care, most notably young and premature patients, suffer from conditions that require the intravenous administration of very potent and short acting pharmaceuticals. These pharmaceuticals typically require a very accurate administration where deviations in flow- and thus dosing-rate can easily result in dosing errors. For this reason, infusion or syringe pumps are used.
On top of this, vascular access to the patient is typically limited and therefore many infusion pumps have to co-administer through one catheter (multi-infusion), making the entire pharmaceutical delivery process complex and hard to predict.
Because dosing errors are common in clinical practice, it was clear that more research was required. Many of the results of this research can be found in the PhD-thesis: “Physical Causes of Dosing Errors in Patients Receiving Multi-Infusion Therapy”.
We conducted a large amount of measurements to learn more about the flow characteristics of multi-infusion setups. These measurements were conducted using Bronkhorst Coriolis flow meters (series mini CORI-FLOW). These flow meters allowed us to measure the flow rate of infusion pumps very accurately, precisely and independent of the density of the solution being measured (although most of the solutions were similar to water).
The flow meters were also chosen because of the suitability for very low flow rates, infusion flow rates may be as low as 0.1 ml/h. Ultimately it is, of course, the dose rate or mass flow rate of the pharmaceutical administered to the patient that is important.
To measure this we used an absorption spectrophotometric setup, which enabled us to measure the concentration of a substance in a solution, i.e. a pharmaceutical or pharmaceutical-analogue. To convert density (e.g. µg/l) to a mass flow rate (e.g. µg/h), the cumulative flow rate (e.g. ml/h) of the infusion setup has to be measured as well.
First we used a precision balance for this but later in the research project we used the mini CORI-FLOW flow meter. The data from the precision balance was rather noisy, whereas the flow meter provided very clean data, which improved our measurements substantially.
However, one point of caution that has to considered is that flow meters do produce a pressure drop resulting in intrinsic flow resistance. The implications of this and how the measurement setup relates to a clinical situation is thoroughly explained in the PhD-thesis.
The research concluded that a wide variety of infusion components all had a particular, usually significant influence and, importantly, medical personnel is usually not aware of the implications this has for infusion therapy. Awareness of the underlying mechanisms of these effects through education and technical innovation were recommended. The Coriolis flow meters from Bronkhorst proved to be very suitable for gaining insight in the different mechanisms of infusion pump system failure.
Further reading: R.A. Snijder - Physical causes of dosing errors in patients receiving multi-infusion therapy (ISBN: 978-94-028-0382-2)
Want to learn more about calibration of infusion pumps? Read the blog of Marcel Katerberg, explaining the calibration techniques to improve infusion pump performance.
Bronkhorst High-Tech designs and manufactures innovative instruments and subsystems for low-flow measurement and control for use in laboratories, machinery and industry. Driven by a strong sense of sustainability and with many years of experience, we offer an extensive range of (mass) flow meters and controllers for gases and liquids, based on thermal, Coriolis and ultrasonic measuring principles. Our global sales and service network provides local support in more than 40 countries. Discover Bronkhorst®!