Bronkhorst propose la plus large gamme de débitmètres et de régulateurs de débit (massique, à faible débit) et de pression du marché. Pour les applications dans les laboratoires, les machines, l'industrie et les zones explosives, de nombreuses versions d'instruments standard et sur mesure peuvent être proposées.
La satisfaction des clients, l'innovation, la qualité des produits et des services sont les clés du succès de Bronkhorst. En se basant sur notre expérience, nos innovations et notre sens des responsabilités, une collaboration avec Bronkhorst assure une Performance for Life.
Les instruments Bronkhorst sont utilisés pour de nombreux marchés et une multitude d'applications. Dans cette rubrique, vous aurez une vue d'ensemble des principaux marchés pour nos équipements, illustré par quelques exemples typiques d'applications.
Vous recherchez une documentation technique, vous souhaitez en savoir plus sur les principes de mesure des instruments Bronkhorst ou souhaitez contacter un ingénieur de service Bronkhorst ? Cette rubrique vous guidera vers les sujets pertinents de service et d'assistance.
Bronkhorst High-Tech BV - leader des technologies de régulateurs et débitmètres massiques pour les gaz et les liquides, des régulateurs de pression et des systèmes d'évaporation.
In basic cancer research, a main aim is to unravel differences between normal cells and cancer cells in order to exploit these differences in the hunt for specific cancer vulnerabilities, so that ways can be found to attack cancer cells while leaving normal cells unaffected.
Since living cells reveal much more information than dead ones, at the Netherlands Cancer Institute (NKI) advanced microscopy techniques are used to extract information about proteins and their interactions in single living cancer cells. To investigate these living cells in their ‘natural habitat’, Bronkhorst was asked to deliver devices for supplying water vapour and gases such as nitrogen, oxygen and carbon dioxide to mimic the conditions in our body.
A dedicated incubator needs to be established that houses the living cells and that ‘keeps them alive’ under conditions that resemble the various oxygen tensions and other conditions encountered in the human body, while being able to be observed by the microscope.
To this end, the moisturised atmosphere in the incubator needs to be controlled accurately at a temperature of 37 °C, should contain at least 5% CO2, and the oxygen content must be adjusted between 2% and 20%.
Bronkhorst delivered three EL-FLOW Select mass flow controllers with read-out units for the controlled supply of air, nitrogen and carbon dioxide gases. By adjusting the relative flows of these gases, an atmosphere was created that contained 2-19 % O2, 0-20 % CO2 and 80-100% N2. This range of gases was used in the incubator to investigate the behaviour of cancer cells, for example to investigate how cancer cells migrate during metastasis or to explore how cells use chemical signals to communicate with each other. Investigations at low oxygen content are especially relevant, as solid tumours are well known to be hypoxic - i.e. they contain less than a few % of oxygen - and this completely alters the physiology of the cells, as well as their response to anti-cancer drugs. All experiments conducted under these controlled conditions showed consistent results.
An artefact of the relatively high operational temperature of 37 °C was that the DMEM medium, a salt solution used for supporting the growth of the living cells, evaporated rather fast. To replenish the evaporated water from the medium, a Bronkhorst μ-FLOW mass flow controller for liquids was used to supply a very stable flow of deionised water. Combined with a local BRIGHT controller with PiPS (Plug-in Power Supply), the μ-FLOW device supplied a controlled water flow between 0.5 and 9.6 microliter per minute.
Empirically, the Netherlands Cancer Institute found that a value of 1.3 μl/min completely compensated for evaporation. Since then, they have been capable of keeping cells alive for weeks while observing them by microscopy. Using this configuration, the relative humidity was kept below 60%, which was necessary to avoid condensation that might damage the sensitive electronics in the setup.