We offer the widest product range of low-flow (mass) flow meters and controllers on the market. Numerous styles of both standard and bespoke instruments can be offered for applications in laboratory, machinery, industry and hazardous areas.
Customer satisfaction, innovation and quality of product and service have been the cornerstones of Bronkhorst's success. Based on our experience, innovation and sense of responsibility, a relationship with Bronkhorst assures Performance for Life.
Bronkhorst instruments are used for numerous applications in many different markets. In this section you will find an overview of the main markets for our equipment, illustrated with some typical examples of applications.
Are you looking for technical documentation, are you interested to learn more about the measuring principles of Bronkhorst products, or you do want to get in contact with a Bronkhorst Service Engineer? This section will guide you to the relevant service & support topics.
Bronkhorst UK is leader in Mass Flow Meter / Mass Flow Controller technology for gases and liquids, Pressure Controllers and Evaporation Systems.
Nitrobenzene is a pale-yellow coloured organic compound with chemical formula C6H5NO2. This liquid compound is mainly used as a precursor for the synthesis of the organic compound aniline, which, in turn, is a precursor for many industrial compounds.
Liquid nitrobenzene needs to be dosed with high accuracy in a chemical pilot plant in order to generate an optimised chemical end product. Before Bronkhorst became involved, the liquid was manually dosed with needle valves. There were however some issues with this procedure. The needle valves were not accurate enough, due to changes in temperature or pressure. The R&D engineers and operators of the pilot plant looked for a better solution, and arrived at Bronkhorst. After observing the mini CORI-FLOW demo case with flow software included, they decided to build a new dosing system with Bronkhorst equipment incorporated.
The reproducibility of the dosing solution for nitrobenzene should be better than ±0.5% of the reading. Furthermore, the dosing solution should compensate for variations in the process upstream of this solution, in order to obtain a stable flow of nitrobenzene.
The new nitrobenzene dosing system comprises a Tuthill gear pump, a degassing valve, three Coriolis mass flow controllers (M13 mini CORI-FLOW series) with filters and some additional equipment. The degassing valve between the gear pump and the mass flow controllers is necessary to tackle the problem of gas bubbles in the nitrobenzene liquid.
After successfully implementing this new dosing system, a repeatability of the measurement value for each (M13) mass flow controller of ±0.1% of the reading has been obtained, well within the requirements. The pilot plant engineers and operators are highly satisfied with this solution.
However, prior to a successful implementation of the Coriolis mass flow controller, some conditions had to be fulfilled. At first, a vibration dampener has to be mounted below the mass blocks of each flow controller. Without such a dampener, an incorrect zero point is obtained. In addition to that these mass blocks were heated, to eliminate potential inaccuracies caused by temperature fluctuations. Furthermore, the pipe between the gear pump and the flow controllers has to be relatively short and straight. Too long, tortuous and too narrow pipes will result in pulsation and gas bubbles in the mini CORI-FLOW mass flow controllers.
Stay up to date with the latest developments in high tech with our newsletter.