We offer the widest product range of low-flow (mass) flow meters and controllers on the market. Numerous styles of both standard and bespoke instruments can be offered for applications in laboratory, machinery, industry and hazardous areas.
Customer satisfaction, innovation and quality of product and service have been the cornerstones of Bronkhorst's success. Based on our experience, innovation and sense of responsibility, a relationship with Bronkhorst assures Performance for Life.
Bronkhorst instruments are used for numerous applications in many different markets. In this section you will find an overview of the main markets for our equipment, illustrated with some typical examples of applications.
Are you looking for technical documentation, are you interested to learn more about the measuring principles of Bronkhorst products, or you do want to get in contact with a Bronkhorst Service Engineer? This section will guide you to the relevant service & support topics.
Bronkhorst High-Tech BV the leaders in Mass Flow Meter / Mass Flow Controller technology for gases and liquids, Pressure Controllers and Evaporation Systems.
A Coriolis mass flow meter is known as a very accurate instrument and it has many benefits compared to other measuring devices. However, every measuring principle has its challenges, as also the Coriolis principle. It can be a real challenge using Coriolis instruments in low flow applications in the heavy industry where you may have to deal with all kinds of vibrations. In this blog I would like to share my experiences with you regarding this topic.
Coriolis mass flow meters offer many benefits above other measuring devices. First of all Coriolis flow instruments measure direct mass flow. This is an important feature for the industry as it eliminates inaccuracies caused by the physical properties of the fluid. Besides this benefit, Coriolis instruments are very accurate, have a high repeatability, have no moving mechanical parts and have a high dynamic range, etc.
In industrial applications, all kinds of vibrations with different amplitudes are very common. A Coriolis meter measures a mass flow using a vibrating sensor tube, which fluctuation gets intentionally out of phase when the fluid flows through. As explained in the video at the end of this article.
This measurement technique is somewhat sensitive to unwanted vibrations with a frequency close to the resonance frequency of the sensor tube (this depends on the sensor tube design, e.g. 360 Hz) or a higher harmonic of this frequency (see picture below).
The likelihood of the occurrence of these unwanted vibrations is higher in an industrial environment. Coriolis flow meter manufacturers do their utmost to reduce the influence of vibrations on the measured value by use of common technical solutions, such as using:
So yes, vibrations can influence the measuring accuracy of your Coriolis flow meter, but only if the vibrations have a frequency close to the resonance frequency. What can you do about this? This depends on the kind of vibration.
In an industry zone frequencies can be generated by:
A well performing flow meter and controller will give the best process result. Therefore, it is advisable to test a Coriolis flow meter in your application if you expect heavy industrial vibrations before you trust it to the full extent. Be careful when filtering the measuring signal. In some cases it makes sense (e.g. when a quick response isn’t required), but if you want to test the performance of a flow meter, filtering could blur your judgement.
If in specific circumstances the Coriolis flow meter isn’t performing the way it should, the operator will see a shift in the process output – for example in an application dosing colour to a detergent it can result in differences in product colour by incorrect dosing and/or unexpected measuring signal behaviour. In these cases it makes sense to check the raw measuring signal (without filters!), because it will give you a good insight in the performance of the flow meter. Ask your flow meter manufacturer how to switch off all signal filtering.