We offer the widest product range of low-flow (mass) flow meters and controllers on the market. Numerous styles of both standard and bespoke instruments can be offered for applications in laboratory, machinery, industry and hazardous areas.
Customer satisfaction, innovation and quality of product and service have been the cornerstones of Bronkhorst's success. Based on our experience, innovation and sense of responsibility, a relationship with Bronkhorst assures Performance for Life.
Learn more about how Coriolis-based flow meters can be used as microfluidic flow meter instead of thermal flow meters. Read the customer story of a microfluidics system builder who used Bronkhorst products.
Are you looking for technical documentation, are you interested to learn more about the measuring principles of Bronkhorst products, or you do want to get in contact with a Bronkhorst Service Engineer? This section will guide you to the relevant service & support topics.
Bronkhorst High-Tech BV the leaders in Mass Flow Meter / Mass Flow Controller technology for gases and liquids, Pressure Controllers and Evaporation Systems.
‘Buy cheap, buy twice’. This is never more apparent than when purchasing new flow meters or looking at ways to protect existing ones if necessary. There are various accessories that you can add-on flow meters, one of them being an in-line filter. In this blog you can read about why use in-line filters and how they work.
In-line filters as add-on
With flow instruments, the accessory of choice often is a communication cable. These are essential in allowing you to communicate with an instrument and monitor or access the very information that you require. However, in-line filters can be far more essential to the long-term performance and life-time of your gas flow instrument.
In-line filters are an essential add-on for flow instruments, particularly in industrial applications where contamination can lead to inaccurate readings, reduced performance or damage to the instrument. Although you might check your research application as well. You may think that in research applications you work with clean gases, but tiny particles can also occur here. Not only the particles in gases can be a problem, the dirt stored in the piping can be harmful as well.
The importance of keeping the gas entering the flow instrument dry and clean cannot be overstated. An in-line filter will protect your gas flow instrument from:
• Debris from contaminated gas lines
• Particles that exist in industrial gases
• Small amounts of oil from compressors
So, do you work with ‘dirty’ gases in your process or gases with particles? Whether this is in an industrial environment or research application… the in-line filter will be your accessory of choice.
The purpose of the in-line filter is to filter the gas before entering the flow meter and to make sure that the gas is clean - without any particles. This way you avoid contamination which can lead to a number of avoidable costs. With further costs I mean costs due to down-time, service costs, calibration costs or time to remove and re-install the damaged instrument.
Besides avoiding contamination, a dry & clean gas is also crucial to ensure a better MTBF (Mean Time Between Failure) of your flow instrument. This is particularly true for flow instruments that use a bypass sensor.
The Bronkhorst flow instruments are designed for low flows and therefore have delicate and finely machined parts. This is needed to enable us to quantify the flow rates of gases that we can achieve with a good level of accuracy and repeatability.
When you consider the potential damage that is possible from the different sources of contamination, and the delicate nature of the internal working of a flow instrument, it would appear to be a very straightforward decision to include a filter in your next purchase of a flow instrument.
Depending on the construction, a flow instrument for gas can be more or less sensitive to contamination. Gas flow instruments can be divided into two sensor principles:
• Gas flow meters using the bypass principle
• Gas flow instrument using the ‘trough-flow’ or CTA principle (Constant Temperature Anemometry)
To learn more about the differences between these two principles, please check our knowledge base.
Within our portfolio Bronkhorst has the IN-LINE filter available for your gas Flow instrument. These IN-LINE gas filters are easy to use. You just screw them into the inlet of the flow instrument, and it guarantees a clean gas inlet. It contains a 316L sintered metal filter cartridge that is suitable for general purpose filtration and can be cleaned with either a suitable solvent or by replacing the cartridge if heavily soiled.
If you use gas that contains large particles, we advise to use a pre-filter. A pre-filter will remove the main part of heavy particles, before they reach the main filter.
When you install a flow meter you want the best performance right away. Check out our top 10 tips for installing your flow meter.
What is the difference between laminar and turbulent flow? 3 Tips on how to minimize the effects of turbulent flow on your flow meters. Learn more!
The choice of piping is important for your flow meter. Check out why and read the tips of our Field Engineer.
Bronkhorst High-Tech designs and manufactures innovative instruments and subsystems for low-flow measurement and control for use in laboratories, machinery and industry. Driven by a strong sense of sustainability and with many years of experience, we offer an extensive range of (mass) flow meters and controllers for gases and liquids, based on thermal, Coriolis and ultrasonic measuring principles. Our global sales and service network provides local support in more than 40 countries. Discover Bronkhorst®!