We offer the widest product range of low-flow (mass) flow meters and controllers on the market. Numerous styles of both standard and bespoke instruments can be offered for applications in laboratory, machinery, industry and hazardous areas.
Customer satisfaction, innovation and quality of product and service have been the cornerstones of Bronkhorst's success. Based on our experience, innovation and sense of responsibility, a relationship with Bronkhorst assures Performance for Life.
Learn more about how Coriolis-based flow meters can be used as microfluidic flow meter instead of thermal flow meters. Read the customer story of a microfluidics system builder who used Bronkhorst products.
Are you looking for technical documentation, are you interested to learn more about the measuring principles of Bronkhorst products, or you do want to get in contact with a Bronkhorst Service Engineer? This section will guide you to the relevant service & support topics.
Bronkhorst High-Tech BV the leaders in Mass Flow Meter / Mass Flow Controller technology for gases and liquids, Pressure Controllers and Evaporation Systems.
Turndown ratio is also commonly referred to as rangeability. It indicates the range in which a flow meter or controller can accurately measure the fluid. In other words, it's simply the high end of a measurement range compared to the low end, expressed in a ratio and is calculated using a simple formula:
Turndown Ratio = maximum flow / minimum flow
For example, if a given flow meter has a 50:1 turndown ratio the flow meter is capable of accurately measuring down to 1/50th of the maximum flow. So, suppose a flow meter has a full scale rating of 20 ln/min the flow meter will measure down to 0.4 ln/min of flow. Keep in mind that the maximum and minimum flow capability of a meter or controller is likely to be a greater span than the measurable and controllable range. For example, a mass flow controller with a 50:1 turndown ratio may have the capability of measuring as high as 25 ln/min or as low as 0.16 ln/min but the turndown ratio will govern the actual measurable range. In this example if the calibrated high flow is 25 ln/min, then the lowest that can be measured is 0.5 ln/min (1/50th of 25). If the application requires that the calibrated minimum flow is 0.1 ln/min, then the maximum flow that can be measured is 5 ln/min (50 times 0.1).